S S Desai
Articles written in Pramana – Journal of Physics
Volume 63 Issue 2 August 2004 pp 465-470
A two-dimensional position sensitive neutron detector has been developed. The detector is a3He + Kr filled multiwire proportional counter with charge division position readout and has a sensitive area of 345 mm × 345 mm, pixel size 5 mm × 5 mm, active depth 25 mm and is designed for efficiency of 70% for 4 Å neutrons. The detector is tested with 0.5 bar3He + 1.5 bar krypton gas mixture in active chamber and 2 bar4He in compensating chamber. The pulse height spectrum recorded at an anode potential of 2000 V shows energy resolution of ∼25% for the 764 keV peak. A spatial resolution of 8 mm × 6 mm is achieved. The detector is suitable for SANS studies in the
Volume 63 Issue 2 August 2004 pp 471-475
A gas-filled microstrip detector for thermal neutrons has been built and successfully tested in our laboratory. The detector has an active area of 20 mm × 15 mm and consists of alternate anodes and cathodes of widths 12 μm and 300 μm respectively. The anode to cathode gap is 150 μm and the pitch is 612 μm. A high resistance, meandering type horizontal strip connects the anodes at one end and aids in position sensing by charge division method. The detector is tested with gas mixtures3He+Kr (1: 2) and3He+CF4 (2:1) at pressure of 3 atmospheres and using a Pu-Be neutron source. The pulse height spectrum shows energy resolution of ∼8% (FWHM) for the 764 keV peak at anode voltage of 525 V for3He+Kr and ∼15% at anode voltage of 800 V for3He+CF4. Gas gains up to 6.3 × 103 and 3.6 × 103 are obtained respectively with these gas mixtures. The overall efficiency of the detector along the sensitive length is tested by exposing the active area to neutrons and recording the position spectrum. The detector shows fairly uniform efficiency (∼45%) over the active length.
Volume 71 Issue 5 November 2008 pp 1183-1188 Neutron Instrumentation
The powder diffractometer and Hi-Q diffractometer at Dhruva reactor make use of five identical 1-D position sensitive detectors (PSDs) to scan scattering angles in the range 3° to 140°. In order to improve the overall throughput of these spectrometers, it is planned to install a bank of 15 high-efficiency and high-resolution PSDs arranged in three layers with five PSDs in each layer. With each high pressure PSD (3He 10 bar + Kr 2 bar) showing the efficiency gain of 1.8 at 1.2 Å, detector bank is expected to show overall gain of 5.5 times the present detection efficiency and reduction in data collection time by equivalent factor. The 1-D PSDs are developed in batches of five, and are characterized so that all PSDs operate at uniform parameters such as position resolution, uniformity of efficiency and linearity of response. Position spectrum indicates the differential position resolution to be $\sim 1$ mm and integral position resolution to be 3–4 mm. Broadening of position spectrum at the extreme end of sensitive length of PSD is analysed using fine shift of the beam. Dependence of position resolution and dynamic range of output pulse on the input impedance of pre-amplifier is also presented.
Volume 93 | Issue 5
Click here for Editorial Note on CAP Mode