R P Shukla
Articles written in Pramana – Journal of Physics
Volume 22 Issue 3-4 March 1984 pp 213-220 Instrumentation
The present paper considers the magnifying power and aberration properties of three different catadioptric magnifying systems useful for application as simple hand-held microfiche readers. These are in the form of a plano-convex lens, a meniscus lens and an achromatic doublet lens and all their outer surfaces have been coated with a semi-reflecting film so that they behave like catadioptric elements. These can be designed to give magnification in the range of 15 × to 25 × with sufficient eye relief. Using such magnifiers, it is possible to read microfiche having frame sizes of 16 × 12 mm and 12 × 9 mm. The cemented achromatic doublet coated on the outer surfaces seems to give a very good image for use in microfiche reader.
Volume 67 Issue 2 August 2006 pp 277-289 Research Articles
An indigenously designed and developed micro-Raman spectrograph, consisting of a diode-pumped solid-state green laser for the excitation of Raman scattering, a Raman imaging microscope, CCD as a detector and a notch filter, has been extensively studied to evaluate its performance. A dielectric edge filter (having 27 alternate layers of SiO2 and TiO2) and a holographic notch filter (Oriel make) have been used to block the Rayleigh scattered light from the sample to the entrance slit of the spectrograph. Holographic notch filter is found to be able to record the Raman shifts below 700 cm−1 conveniently whereas dielectric edge filter (27 layers) has enabled the spectrograph to record the Raman spectra very efficiently after a wave-number shift of 700 cm−1. It has also been observed that the instrument using the edge filter provides a peculiar spectrum consisting of three spectral lines having Raman shifts as 569, 1328 and 1393 cm−1 in the Raman spectrum of a weakly scattering sample with large reflectivity. Similarly, a spectrum consisting of multiple lines has been observed when the instrument is being operated using a holographic notch filter. These spectral lines are not observed in the case of liquid samples such as benzene, carbon tetrachloride, ethanol, diethyl ether etc. The origin of these peculiar spectral lines has been briefly discussed in the paper. Additionally, a major motivation for this work is to utilize the results for the selection of an appropriate filter depending on the type of the sample, i.e. weakly scattered and highly reflecting sample or highly scattered and low reflecting sample.
Volume 94, 2020
Continuous Article Publishing mode
Click here for Editorial Note on CAP Mode