J N Soni
Articles written in Pramana – Journal of Physics
Volume 10 Issue 3 March 1978 pp 289-302 Experimental Techniques And Instrumentation
The design and fabrication of an indigenous on-line computer controlled four-circle neutron diffractometer at the CIRUS reactor in Trombay are described. The diffractometer has an 18 in dia full-circle crystal-orienter which is sturdy enough to carry a cryostat. Three crystal orientation angles χ, φ and ω and the detector angle 2θ can be set to an accuracy of 0·01°. The four angle shafts are driven through precision worm-gears by SCR-controlled DC motors and their instantaneous positions sensed by optical digitizers. The diffractometer is interfaced to an ECIL TDC-312 computer system consisting of the CPU with 4K-memory, ASR-35 teletype, X-Y plotter and the digital input/output system (DIOS). The DIOS which operates under program control is a real-time peripheral device used to exchange information in digital form between the computer and the diffractometer. A software package consisting of over 40 useroriented teletype commands has been developed for on-line control and automatic data-acquisition.
Volume 19 Issue 4 October 1982 pp 323-336 Experimental Techniques And Instrumentation
Design and construction details of a horizontal 2 MV Tandem Van de Graaff accelerator built at Bhabha Atomic Research Centre are given. A terminal voltage of 2.15 MV has been achieved. Energy analysed Corona stabilized beams of protons and oxygen ions have been obtained. Experiments have been carried out to test the performance of the accelerator.
Volume 96, 2022
Continuous Article Publishing mode
Click here for Editorial Note on CAP Mode