Articles written in Pramana – Journal of Physics
Volume 71 Issue 5 November 2008 pp 1085-1089 Neutron Optics and Reflectometry
Grazing incidence neutron reflectometry provides an opportunity to measure the depth profile of a thin film sample with a resolution <1 nm, in a non-destructive way. In this way the diffusion across the interfaces can also be measured. In addition, neutrons have contrast among the isotopes, making it feasible to measure the self-diffusion. In the present work, the isotope multilayers of [FeN/57 FeN]10 and [FeN/Fe15N]10 were prepared using magnetron sputtering and self-diffusion of Fe and N was investigated. It was found that N diffusion is slower compared to Fe and does not follow the atomic size dependence.
Volume 83 Issue 5 November 2014 pp 807-815
P Sugathan A Jhingan K S Golda T Varughese S Venkataramanan N Saneesh V V Satyanarayana S K Suman J Antony Ruby Shanti K Singh S K Saini A Gupta A Kothari P Barua Rajesh Kumar J Zacharias R P Singh B R Behera S K Mandal I M Govil R K Bhowmik
The characteristics and performance of the newly commissioned neutron detector array at IUAC are described. The array consists of 100 BC501 liquid scintillators mounted in a semispherical geometry and are kept at a distance of 175 cm from the reaction point. Each detector is a $5''\times 5''$ cylindrical cell coupled to $5''$ diameter photomultiplier tube (PMT). Signal processing is realized using custom-designed home-made integrated electronic modules which perform neutron–gamma discrimination using zero cross timing and time-of-flight (TOF) technique. Compact custom-built high voltage power supply developed using DC–DC converters are used to bias the detector. The neutrons are recorded in coincidence with fission fragments which are detected using multi-wire proportional counters mounted inside a 1m diameter SS target chamber. The detectors and electronics have been tested off-line using radioactive sources and the results are presented.
Volume 94, 2020
Continuous Article Publishing mode
Click here for Editorial Note on CAP Mode