Y. D. Mayya
Articles written in Journal of Astrophysics and Astronomy
Volume 12 Issue 4 December 1991 pp 319-331
Calibration of the CCD camera of the 1m telescope at the Vainu Bappu Observatory, Kavalur, to the
Volume 13 Issue 1 March 1992 pp 129-144
The system gain of two CCD systems in regular use at the Vainu Bappu Observatory, Kavalur, is determined at a few gain settings. The procedure used for the determination of system gain and base-level noise is described in detail. The Photometrics CCD system at the 1-m reflector uses a Thomson-CSF TH 7882 CDA chip coated for increased ultraviolet sensitivity. The gain is programme-selected through the parameter ‘cgain’ varying between 0 and 4095 in steps of 1. The inverse system gain for this system varies almost linearly from 27.7 electrons DN-1 at cgain = 0 to 1.5 electrons DN-1 at cgain = 500. The readout noise is ≲ 11 electrons at cgain = 66. The Astromed CCD system at 2.3-m Vainu Bappu Telescope uses a GEC P8603 chip which is also coated for enhanced ultraviolet sensitivity. The amplifier gain is selected in discrete steps using switches in the controller. The inverse system gain is 4.15 electrons DN-1 at the gain setting of 9.2, and the readout noise ∼ 8 electrons.
Volume 43, 2022
Continuous Article Publishing mode
Since January 2016, the Journal of Astrophysics and Astronomy has moved to Continuous Article Publishing (CAP) mode. This means that each accepted article is being published immediately online with DOI and article citation ID with starting page number 1. Articles are also visible in Web of Science immediately. All these have helped shorten the publication time and have improved the visibility of the articles.
Click here for Editorial Note on CAP Mode