Articles written in Journal of Astrophysics and Astronomy
Volume 38 Issue 2 June 2017 Article ID 0032 Review Article
M. C. Ramadevi B. T. Ravishankar N. Sitaramamurthy G. Meena Brajpal Singh Anand Jain Reena Yadav Anil Agarwal V. Chandra Babu Kumar Ankur Kushwaha S. Vaishali Nirmal Kumar Iyer Anuj Nandi Girish V. Vivek Kumar Agarwal S. Seetha Dipankar Bhattacharya K. Balaji Manoj Kumar Prashanth Kulshresta
We report the in-orbit performance of Scanning Sky Monitor (SSM) onboard AstroSat. The SSM operates in the energy range 2.5 to 10 keV and scans the sky to detect and locate transient X-ray sources. This information of any interesting phenomenon in the X-ray sky as observed by SSM is provided to the astronomical community for follow-up observations. Following the launch of AstroSat on 28th September, 2015, SSM was commissioned on October 12th, 2015. The first power ON of the instrument was with the standard X-ray source, Crab in the field-of-view. The first orbit data revealed the basic expected performance of one of the detectors of SSM, SSM1. Following this in the subsequent orbits, the other detectors were also powered ON to find them perform in good health. Quick checks of the data from the first few orbits revealed that the instrument performed with the expected angular resolution of 12’ × 2.5∘ and effective area in the energy range of interest. This paper discusses the instrument aspects along with few on-board results immediately after power ON.
Volume 42 All articles Published: 3 July 2021 Article ID 0070 PAYLOAD CALIBRATION
SSM onboard AstroSat is designed to monitor X-ray sky in the energy range 2.5–10 keV to detect and locate X-ray sources in outburst. SSM with its three almost identical 1D-proportional counters mounted on a rotating platform, scans the sky in step and stare mode of operation. It observes the X-ray skyand generates light curves for X-ray sources detected. Here, we discuss the positional calibration to carry out imaging with SSM. Onboard calibration of SSM has been carried out with Crab, the standard X-ray source. SSM observations of Crab are compared with that of MAXI on ISS for cross calibration of the instrument.
Volume 43, 2022
Continuous Article Publishing mode
Since January 2016, the Journal of Astrophysics and Astronomy has moved to Continuous Article Publishing (CAP) mode. This means that each accepted article is being published immediately online with DOI and article citation ID with starting page number 1. Articles are also visible in Web of Science immediately. All these have helped shorten the publication time and have improved the visibility of the articles.
Click here for Editorial Note on CAP Mode