Arsenic can be determined in parts-per-million (ppm) level by absorbance measurement. This method is based on the quantitative colour bleaching of the dye, methylene blue by arsine catalyzed by nanoparticles in micellar medium. The arsine has been generated in situ from sodium arsenate by NaBH4 reduction. The absorbance measurement was carried out at the 𝜆max of the dye at 660 nm. The calibration graph set-up for three linear dynamic ranges (LDR) are 0–8.63 ppm, 0–1.11 ppm and 0–0.11 ppm and limit of detections (LODs) are 1.3, 0.53 and 0.03 ppm, respectively. This method is simple, sensitive and easy to carry out. It is free from phosphate and silicate interference and applicable to real sample analysis.
Volume 45, 2022
Continuous Article Publishing mode
Prof. Subi Jacob George — Jawaharlal Nehru Centre for Advanced Scientific Research, Jakkur, Bengaluru
Chemical Sciences 2020
Prof. Surajit Dhara — School of Physics, University of Hyderabad, Hyderabad
Physical Sciences 2020
Click here for Editorial Note on CAP Mode